View and Download Dräger Polytron instructions manual online. Transmitters for electrochemical sensors. Polytron Transmitter pdf manual download. View and Download Dräger Polytron instructions for use manual online. ( approved as type P3S). Transmitter for electrochemical Sensors. Polytron The Dräger Polytron® is an intrinsically safe gas detector for the continuous The Polytron consists of two components, the docking station, . Please note – Product designs and specifications are subject to change without notice.
|Published (Last):||4 September 2010|
|PDF File Size:||5.56 Mb|
|ePub File Size:||4.77 Mb|
|Price:||Free* [*Free Regsitration Required]|
For Your Safety For Your Safety Strictly follow the Instructions for Use Any use of the apparatus requires full understanding and strict observation of these instructions.
The apparatus is only to be used for purposes specified here. Maintenance The unit must be inspected and serviced regularly by suitably qualified persons. Repair and general overhaul of the apparatus may only be carried out by trained service personnel.
The Polytron transmitter may also be connected to other central units if the 30000 conditions are met: NOTE In explosion-hazard areas: Observe the national regulations concerning electrical equipment in explosion-hazard areas. Installing the transmitter Installing the docking station — If the transmitter is to be installed in a Zone 2 explosion-hazard area, select a location with low exposure to mechanical risk.
Shorten the shield if installed to prevent short-circuiting: Cut excess wires short or 2 Fasten in 4-pin terminal. Remove sensor from packaging.
DRäGER POLYTRON INSTRUCTIONS FOR USE MANUAL Pdf Download.
This current is proportional to the gas concentration. Maintenance Maintenance Maintenance intervals Before starting operation: Check the calibration, see page Check the transmission of signals to the control unit and the triggering of alarms. Maintenance Calibrating the zero point For all sensors except oxygen sensor: The zero polyteon can be calibrated without the use of nitrogen zero gas when the ambient air is free from measuring gas and other interfering gases.
Polytron 1 Use the calibration adapter. Set maintenance switch to maintenance position, see page Maintenance Replacing the rdager The sensor can be replaced, if necessary, without interrupting the power supply in the explosion-hazard area. Display Equipment fault, Only use a sensor with the gas type, Part No. Splash guard 68 07 Splash guard AC sensor 68 09 CSA approval Master Contract: The products listed, including the latest revision described below, are eligible to be marked in accordance with the referenced Certificate.
Dräger Polytron 3000 Instructions For Use Manual
Also See for Polytron Instructions for use manual – 52 pages. Page of 68 Go.
Page 9 – Installing the transmitter in areas subj Page 10 – Installing the transmitters in non-explo Table of Contents Add to my manuals Add. Table Of Contents Order List Installing The Docking Station Installing the transmitter 3000 the docking station — If the transmitter is to be installed in a Zone 2 explosion-hazard area, select a location with low exposure to mechanical risk. Fitting The Sensor approx.
Start-up — A current between 4 and 20 mA flows through the transmitter during normal ope- ration. Maintenance Maintenance Maintenance Maintenance intervals Before starting operation: Replacing The Sensor Maintenance Replacing the sensor The sensor can be replaced, if necessary, without interrupting the power supply in the explosion-hazard area.
Page 23 Order No. Page 24 Order No. Page 26 ATEX approval Page 27 ATEX approval Page 28 ATEX approval Page 29 ATEX approval Page 30 ATEX approval Page 31 ATEX approval Page 32 ATEX approval Page 33 ATEX approval Page 34 ATEX approval Page 35 ATEX approval Page 36 ATEX approval Page 37 ATEX approval Page 38 ATEX approval Page 39 ATEX approval Page 40 ATEX approval Page 41 ATEX approval Page 42 ATEX approval Page 43 ATEX approval Page 44 ATEX approval Page 45 ATEX approval Page 46 ATEX approval Page 48 IECEx approval Page 49 IECEx approval Page 50 IECEx approval Page 52 UL approval Page 58 CSA approval Page 61 CSA approval Page 63 Declaration of Conformity Comments to this Manuals Your Name.
Enter text from picture: By pressing ‘print’ button you will print only current page. To print the manual completely, please, download it. Don’t show me this message again.
Got it, continue to print. Don’t have an account?